Keywords: STEMI Kit, NIBP, SpO2, ECG, Electrode, B.P cuff, Mobmon 12.0
| Show Slide
Kallows STEMI Kit.
|Hello and welcome to this tutorial on the Kallows STEMI Kit.|
|Learning Objectives|| In this tutorial we will learn to -
| Show Slide
|To practise this tutorial, you will need the Kallows STEMI Kit.|
|Show Image|| The STEMI Kit consists of
|Show image||This is the Mobmon device.|
|Show image||It has a power button along with the charging port on the left side.|
|Show image||And the SpO2 and ECG ports on the back side.|
|Show image||This cable head has to be inserted into the ECG port on the Mobmon device.|
|Show image||After connecting, secure the connection with the screws provided on either sides.|
|Show image||Next, we will learn about the SpO2 probe.|
|Show image|| This probe is made up of the following parts-
|Show image||Let’s now see how to position the SpO2 probe.|
|Show image||Connect the oximetry probe/cable to the SpO2 connector on the Mobmon device.|
|Show image||This is how it should look when connected.|
|Show image||The patient's finger must be inserted right to the end of the sensor as shown in the figure.|
|Show image|| While selecting a sensor site, priority should be given to an extremity free of -
|Show image|| The typical sites for oximetry probe connections are -
|Show images as per narration|| Pls note:
|NIBP CUFF PLACEMENT||Next, let’s learn how to position Blood Pressure cuff.|
|Show image||Connect the cuff connector to NIBP Connector of the Bluetooth device.|
|Press the Start button to activate Bluetooth BP Monitor.|
|Show the images as per narration|| Pls note:
Inaccurate measurements may be caused by incorrect application or use such as-
|Show the images as per narration|| Do not connect the SpO2 sensor and the B.P cuff to the same limb of the patient.
This is to avoid nuisance alarms.
|Show image||Make sure that while measuring the NIBP, the tube of the cuff is not blocked or tangled.|
|Finally, let’s see how to position the ECG leads.|
|PATIENT PREPARATION||Quality ECG rhythm can be obtained with good skin preparation and proper electrode placement.|
Show images as per narration one by one
| The following must be done for good patient preparation-
|Show image|| Note:
Ensure that conductive portions of the
|Show image|| This is how the electrodes should be placed.
|Show the image||Make sure the power cord and patient cable do not cross each other.|
|Show image||Switch on the Mobmon device by selecting power on/ off button on the side.|
|STEMI App to be shown here||Select ECG tab on the STEMI device to view the ECG, in the ECG live steam page.|
|Summary slide|| Let us summarize.
In this tutorial, we have learnt -
|Show STEMI INDIA slide|| STEMI INDIA
|Show Spoken Tutorial Project Slide|| Spoken Tutorial Project, IIT Bombay is funded by NMEICT, MHRD, Govt. of India.
|Show THANK YOU slide|| This tutorial has been contributed by
This is Nancy Varkey signing off. Thanks for joining.